Intecont Tersus – Free download as PDF File .pdf), Text File .txt) or read for weighing tasks better than % Manual and/or automatic zero setting Full. The INTECONT Tersus weighing elec- tronics is specially tion module, INTECONT Tersus optimally fits into any Manual and/or automatic zero setting. processes, the INTECONT Tersus meas- uring, control, and supervisory module, INTECONT Tersus optimally fits into any Manual and automatic zero setting.
|Published (Last):||22 December 2006|
|PDF File Size:||18.74 Mb|
|ePub File Size:||10.24 Mb|
|Price:||Free* [*Free Regsitration Required]|
The belt speed is normally constant. The hatched area shows the load distribution on the weighing platform. The zero setting records the manuxl point fault on an empty belt over one or several whole-number belt circuits. Minor risks may arise in these situations if the measuring system is used or operated by untrained personnel.
Multiple-Idler Platform Several weighing idlers are combined into one mechanical unit, the measuring bridge. Incorrectly set para-meters may cause injury or material damage. Does not apply for the auto-matic zero setting device. Only personnel familiar with intecoht device’s mode of operation may alter these parameters e.
Manual Intecont Tersus
It can be printed out with the Batch Record function or inteclnt after every finished batch P The feed rate will only be calculated with the feeder switched on and during the stopping time P The following conversion formula applies to the effective bridge length with single-roll weighbridges: Out of Tolerance W1 P Volumetric Synchronous In contrast to volumetric operation, it is not the rated belt load that is used for calculating the set belt speed.
Then find out more about the individual components at www. The result is compared to a predetermined reference value and then displayed. There is danger of contacting live parts. Furthermore they may also cause considerable disruption to weigh operations. PersonnelPersonnelPersonnelPersonnel Preparation, assembly, commissioning, tesus, maintenance and servicing may only be carried out by qualified personnel.
The batch protocol can only be printed after completing the batch. The measuring system can be part of a more complex plant.
Damaged or defective electrical components must immediately be re-placed or, if possible, repaired by qualified personnel. The Ethernet network connection is included in the basic equipment. The functions of buttons 4 – 7 depend on the application software and parameterization.
Key in the length ratio in percent in parameters Platform Dis. Calibration active is shown in the header. Intecont Tersus Schenck Calibration Manual pdf The inclination of the belt may only be entered for systems in which the load cells are arranged perpendicular to the belt.
This especially applies to alterations which could affect the operating safety of the system. The complete parameter list is printed out with the current readings using the Print Parameters function. Access to some functions can be configured in the Dialog Behaviour block. This program even makes it easy manuzl non-experts to undertake all the settings and corrections required. Some operating functions can be triggered by various sources. Moisture and HumidityMoisture and HumidityMoisture and HumidityMoisture and Humidity All scales parts, electrical etrsus in particular, must be protected from moisture and humidity when the housing is opened for e.
Selecting the physical output for triggering an external itecont counter for the measured quantity Bear in mind the limitations on speed and life cycle when selecting a relay output and set the weight equivalent for an impulse in P All alarms are also sig-naled through a relay output. The basic diagram of the ITE inputs and outputs It must tresus if the weighing platform is placed under load e.
Call up the function block Calibration Set Time Abort possible at any time Enter the year, month, day, hour, minute and second. The effective platform length and the inclination of the belt must be measured on-site. Special start-up and empty running functions can also be activated.